|About us | Contact | Support
|Laser safety resources and considerations|
|Relevant information and web links for laser safety - using lasers in the US and worldwide|
|Home > Resources > Laser safety|
Acuity’s laser sensors put out about the same power levels as the laser pointers that have become quite common, but they are still subject to safety regulations. These devices are potentially hazardous only when the beam enters the eye directly or through optics such as mirrors or focusing lenses. Scattered light or light striking the skin is not classified as hazardous.
For cw lasers, Class IIIa lasers are visible lasers with output power of more than 1 mW but less than 5 mW, as measured through a 7 millimeter aperture. Class IIIb lasers are those with output above 5 mW, or any laser outside the visible frequency band that is not unconditionally eye safe. Class IIIb extends up to 500 mW output power.
Regulations for light-emitting devices are governed by the U.S. Food and Drug Administration (FDA) under 21 CFR 1040.10, PERFORMANCE STANDARDS FOR LIGHT-EMITTING PRODUCTS .
Classifications of the AR4000 Versions
The AR4000-LV is a Class IIIa laser device, with a maximum power of less than 5 mW. The aperture cover supplied with the 4000 LV is required for end user sales in the U.S. OEMs and developers may integrate a separate aperture cover, so long as it meets Federal requirements.
The AR4000-LIR and high power LIR are a Class IIIb laser devices. The aperture cover supplied with these sensors is required for end user sales in the U.S. In addition, complete systems (with power supply) must also include a keyswitch and power interlock jack. The key must be removable only when the laser is off. The power jack disconnects power to the laser when it is removed. The AR4000-LIR power supply includes the keyswitch and interlock jack. End users providing their own power supplies must include a conforming keyswitch and interlock. A separate keyswitch/interlock box is available for the 4000-LIR.
Laser classification becomes even more complex when considering scanned beams. Depending on the speed of the scan and whether a scan repeats along one line moves in 2 axes, Acuity’s sensors can be made eye safe while scanning. Regulations require end user scanning systems that are classified as eye safe to include interlocks that turn the laser off if the beam scanning speed or pattern changes in any way that could cause exposure to hazardous light levels. Scanning systems may also be classified as Class II, IIIa, or IIIb, but precautions must be taken to assure that they are used in a safe manner.
The ANSI document Z136.1-1993, “American National Standard for Safe Use of Lasers” describes the classifications of lasers and the precautions to be taken for each class. This document may be ordered from ANSI, which has offices in Hackensack, NJ, and New York, NY. (Phone: 212-642-4900)
Helpful links for laser safety
> Are you using a laser device in the United States? Consult the OSHA website for laser safety hazards.
> For Europe, the guiding regulatioin is IEC 60825-1. You can download it from the IEC website.
Laser safety for Acuity rangefinders and senors
|Contact Acuity||Change Language||Sitemap||Company News||FAQ's|
|Laser distance measuring sensors||Laser sensor applications||Rangefinder & laser sensor resources||Local sales contacts||About Acuity|
|© Copyright 2009, Schmitt
Measurement Systems, Inc.
Makers of precision surface measurement and alignment products